Degree Granting Department
Venkat Bhethanabotla, Ph.D.
Robert Frisina, Ph.D.
Byeong Cha, Ph.D.
bioassay, immunofluorescence, Micro-mixing, nanoparticles, removal
Immunofluorescence assays are capable of both detecting the amount of a protein and the location of the protein within a cell or tissue section. Unfortunately, the traditional technique is not capable of detecting concentrations on the nanoscale. Also, the technique suffers from non-specific attachment, which can cause false-positives, as well as photobleaching when detecting lower concentrations is attempted. There is also a time constraint problem since the technique can take from many hours to a few days in some cases.
In this work, metal-enhanced fluorescence (MEF) is used to lower the detection limit and reduce photobleaching. Unfortunately, MEF also increases the intensity of non-specifically bound proteins (NSBPs). Therefore, a surface acoustic wave (SAW) device is used to remove the more weakly bound NSBPs. Previously, this has been shown on lithium niobate, but it is used with a quartz substrate in this work. The SAW device is also used to cause micro-mixing which speeds the process up significantly.
In this research, it was found that silver nanocubes can lower the detection limit down to below 1 ng/mL. Quartz SAW devices are shown to remove NSBPs at a power of 10 mW applied for five minutes. Micro-mixing is shown to be improved by a factor of six at 10 mW for 10 minutes by saturating the antibody used in this research, which takes 1 hour without micro-mixing. Finally, all three components are combined. In this work, the whole device is used to detect 50 ng/mL. After micro-mixing, the intensity is the same as with MEF, and, after removal, it has been lowered by 7 a.u.
Scholar Commons Citation
Morrill, Samuel, "Combined Metal-Enhanced Fluorescence-Surface Acoustic Wave (MEF-SAW) Biosensor" (2014). Graduate Theses and Dissertations.